BY PASS Rotameter for Chemical process Indusrtries
Flowtech Make By-Pass Rotameters.
Make Bypass Rotameter as designed for measurement of flow rate of Liquid and gases in 2 NB or higher Size pipe sine.
Flowtech Make By- pass Rotameter as designed for measurement of flow rate of Liquid and gases in 2 NB or higher Size pipe sine. The complete assembly consist of carrier rings [orifice Flanges], orifice Plate, by-pass piping with Isolation valves & a Rotameter as indicator. The orifice plate is design on IS of BS specification.
These meters are selected for low cost measurement of high flow rates generally in the 2 and larger sizes. It provides linear measurements of fluid in conjunction with an orifice plate in the main line. The Rotameter is modified to include a range orifice which is sized so that the flow through the meter at the maximum pressure drops across the orifice plate in necessary to lift the float to the maximum position. Hence the flow through the range orifice is proportional to the instantaneous flow through the main orifice and so the bypass Rotameter measure main line flow.
Suitable for larger pipeline of 50 NB to 450 NB.
A bypass rotameter, also known as a bypass flow meter, is a type of flow measurement instrument that measures the flow rate of a fluid in a system using a bypass line. It is made up of two parallel lines: a major flow line and a smaller bypass line. Bypass rotameters are widely employed in situations where accurate flow measurement is required but installing a full-sized flow meter in the main flow line is either impossible or prohibitively expensive. They offer a low-cost option for monitoring fluid flow rates in a variety of sectors, including chemical processing, water treatment, and HVAC systems.
Are bypass rotameters costly?
Bypass rotameters are often less expensive than other flow measuring devices.
Is there a low pressure drop with bypass rotameters?
These devices often have a minimal pressure drop, reducing system energy losses.
Are bypass rotameters appropriate for all fluids?
Certain fluids, such as those that are very corrosive or abrasive, may be incompatible with the materials used in bypass rotameters.