Flowtech make Transparent Level Gauges employ two transparent glasses fitted with a liquid chamber on either side. The liquid level is indicated as a result of difference in the transparent properties of the two media. For water / steam applications, an illuminator is mounted on the rear side of the gauge with its light rays deflected upward into the water column. This enables the observer to see illuminated surface of the water as the light rays impinged on the surface of separation between water and steam are reflected back to the eye of the observer.
A part from glass tube level gauges, transparent level gauges are always fitted with two plate transparent glasses between which the fluid is contained. The fluid level is indicated as the result of the different transparency of the two media and in some cases (for water steam), by conveying upwards on to the surface of separation (between liquid and gaseous substances) a source of light located at the back of the gauge, the rays of which are totally reflected down to the observer.
Transparent level gauges are suitable for almost all installations. In fact they permit:
REFLEX LEVEL GAUGE
Flowtech make Reflex Level Gauge is designed for liquid level indication in vessels under high pressure & temperature conditions. These Reflex Level Gauges are widely demanded in various pharmaceuticals and other industries due to their high performance and longer service ability in the domestic as well as international market.
Reflex glass level gauges working principle is based on the light refraction and reflection laws. Reflex glass level gauges use glasses having the face fitted towards the chamber shaped to have prismatic grooves with section angle of 90Ã‚°. When in operation, the chamber is filled with liquid in the lower zone and gases or vapors in the upper zone; the liquid level is distinguished by different brightness of the glass in the liquid and in the gas/vapor zone. The reflex level gauges do not need a specific illumination: the day environmental light is enough. Only during the night an artificial light must be provided.
Reflex glass level gauges can be used in most of the cases and offer great advantages in terms of: low initial cost, low operating cost, easy level reading. Reflex level gauges cannot be used in certain cases as for example: